© 2005-2006 Shenzhen Mindray Bio-medical Electronics Co., Ltd. All rights Reserved. For this Service Manual, the issued Date is 2006-11 (Version: 1.1).
Intellectual Property Statement
SHENZHEN MINDRAY BIO-MEDICAL ELECTRONICS CO., LTD. (hereinafter called Mindray) owns the intellectual property rights to this Mindray productand this manual. This manual may refer to information protected by copyrights or patents and does not convey any license under the patent rights of Mindray, nor the rights of others. Mindray does not assume any liability arising out of any infringements of patents or other rights of third parties. Mindray intends to maintain the contents of this manual as confidential information. Disclosure ofthe information in this manual in any manner whatsoever without the written permission of Mindray is strictly forbidden. Release, amendment, reproduction, distribution, rent, adaption and translation of this manual in any manner whatsoever without the written permission of Mindray is strictly forbidden. , , , , are the registered trademarks or All other trademarks that appear in this
trademarksowned by Mindray in China and other countries. the property of their respective owners.
manual are used only for editorial purposes without the intention of improperly using them. They are
Responsibility on the Manufacturer Party
Contents of this manual are subject to changes without prior notice. All information contained in this manual is believed to be correct. Mindray shall not beliable for errors contained herein nor for incidental or consequential damages in connection with the furnishing, performance, or use of this manual. Mindray is responsible for safety, reliability and performance of this product only in the condition that:
• all installation operations, expansions, changes, modifications and repairs of this product are
conducted by Mindray authorized personnel;• the electrical installation of the relevant room complies with the applicable national and local
• the product is used in accordance with the instructions for use.
Upon request, Mindray may provide, with compensation, necessary circuit diagrams, calibration illustration list and other information to help qualified technician to maintain and repair someparts, which Mindray may define as user serviceable.
Note This equipment must be operated by skilled/trained medical professionals.
Warning For continued safe use of this equipment, it is necessary that the listed instructions are followed. However, instructions listed in this manual in no way supersede established medical practices concerning patient care. Do not rely only on audible alarmsystem to monitor patient. When monitoring adjusting the volume to very low or completely muting the sound may result in the disaster to the patient. The most reliable way of monitoring the patient is at the same time of using monitoring equipment correctly, manual monitoring should be carried out. This multi-parameter patient monitor is intended for use only by medical professionals in health careinstitutions. To avoid electrical shock, you shall not open any cover by yourself. Service must be carried out by qualified personnel. Use of this device may affect ultrasonic imaging system in the presence of the interfering signal on the screen of ultrasonic imaging system. Keep the distance between the monitor and the ultrasonic imaging system as far as possible. It is dangerous to exposeelectrical contact or applicant coupler to normal saline, other liquid or conductive adhesive. Electrical contact and coupler such as cable connector, power supply and parameter module socket-inlet and frame must be kept clean and dry. Once being polluted by liquid, they must be thoroughly dried. If to further remove the pollution, please contact your biomedical department or Mindray. It is...