• Product must be maintained and installed in strict accordance with the National Electrical Codes, Gems technical brochure, instruction bulletin, and any applicable electrical code in the country in which the product is installed. Failure to observe this warning could result in serious injuries or damages. • For hazardous area applications involving such things as (but notlimited to) ignitable mixtures, combustible dust and flammable materials, use an appropriate intrinsically safe interface device. • Warning: To prevent ignition of flammable or combustible atmospheres, disconnect power before servicing. • The pressure and temperature limitations shown on the individual catalog pages and drawings for the specified level sensors must not be exceeded. These pressuresand temperatures take into consideration possible system surge pressures/ temperatures and their frequencies. • Selection of materials for compatibility with the media is critical to the life and operation of GEMS level sensors. Take care in the proper selection of materials of construction, particularly wetted materials. • Electrical entries and mounting points in an enclosed tank may requireliquid/vapor sealing. • Physical damage sustained by the product may render it unserviceable.
Continuous Level Transmitters
XM-800/XMP-800 Analog Output (Proportional Voltage) XT-800/XTP-800 Signal Conditioned Output (4-20mA, 0-5 VDC, 0-12 VDC)
Instruction Bulletin No. 179685
XM-800 Series includes models XM-800, XMP-800, XM-850, XM-820 XT-800 Series includes models XT-800, XTP-800,XT-850, XT-820
Designed for continuous liquid level sensing, GEMS transmitters are considered "components".
The XM/XT-800 utilizes reed switch/magnet technology. A magnet-equipped float rises or lowers with corresponding liquid level. The magnetic field generated from the float
Cancellations and returns are accepted up to30 days from date of order. You must contact our Returns Department for a Return Authorization (RA) number. Then return goods, freight prepaid, in the original container and include original packing slip. C.O.D. returns are not accepted. Gems Sensors reserves the right to apply restocking or cancellation charges.
actuates a series of reed switches mounted within a sealed hollow tube. The seriesof reed switches is combined with resistors to form a voltage divider. When a regulated DC voltage is applied to an XM-800, the
Sealed Hollow Tube
Gems Sensors, the seller, warrants its products to be free from defects in material and workmanship in normal use and service for a period of one year from date of shipment. Gems Sensors reserves the right and option to refundthe purchase price in lieu of repair or replacement upon evaluation of the returned original part. Modification, misuse, attempted repair by others, improper installation or operation shall render this guarantee null and void. Imo Industries Inc., Gems Sensors, makes no warranty of merchantability or fitness for a part or purpose. Limits of Liability: In no circumstances shall Gems Sensors be liablefor special, consequential or exemplary damages of any kind or character, including contract, tort, and strict liability in tort and contract. Equipment sold by Gems Sensors is not intended for use in a nuclear installation, nor shall it be used as a "Basic Component" as same is defined under Part 21, Title 10 of the Code of Federal Regulations. In the event of such use, you agree to indemnifyand hold us harmless from any and all subsequent liabilities and responsibilities which might arise in connection with such use.
resulting voltage output is directly proportional to liquid level. An XT-800 is an XM-800 with a signal conditioned output, for use in applications that require unregulated input voltage or current output.
Installation / Mounting
Units operate normally in any...