• Bedienungsanleitung • Manual de Instrucciones • Manuel d’Utilisation
English O P E R A T O R ’ S M A N U A L
FG2C/FG3C Function Generator
WARRANTY The FG2C and FG3C Function Generators are warranted against any defects of material or workmanship within a period of one (1) year following the date ofpurchase of the Function Generator by the original purchaser or original user. Any Function Generator claimed to be defective during the warranty period should be returned with proof of purchase to a Wavetek authorized Service Center or to the local Wavetek dealer or distributor where the Function Generator was purchased. See maintenance section for details. Any implied warranties arising out of thesale of a Wavetek Function Generator, including but not limited to implied warranties of merchantability and fitness for a particular purpose, are limited in duration to the above stated one (1) year period. Wavetek shall not be liable for loss of use of the Function Generator or other incidental or consequential damages, expenses, or economical loss or for any claim or claims for such damage,expenses or economical loss. Some states do not allow limitations on how long implied warranties last or the exclusion or limitation of incidental or consequential damages, so the above limitations or exclusions may not apply to you. This warranty gives you specific legal rights, and you may also have other rights that vary from state to state. CERTIFICATIONS AND PRECAUTIONS This instrument isEN61010-1:1993 certified for Installation Category II, Pollution Degree II. All inputs are protected against continuous overload conditions up to the limits of each function's stated input protection (see specifications). Never exceed these limits or the ratings marked on the instrument itself. Always inspect your Function Generator, test leads and accessories for signs of damage or abnormality beforeevery use. If an abnormal condition exists (broken or damaged test leads, cracked case, display not reading, etc.), do not use. Never ground yourself when taking measurements. Do not touch exposed metal pipes, outlets, fixtures, etc., which might be at ground potential. Keep your body isolated from ground and never touch exposed wiring, connections, test probe tips, or any live circuit conductors.Do not operate instrument in an explosive atmosphere (flammable gases, fumes, vapor, dust.) Do not use this or any piece of test equipment without proper training. DANGER High Voltage ATTENTION Refer to Manual Protective Conductor Terminal Frame or Chassis Terminal WARNING: To avoid electrical shock, the power cord protective grounding conductor must be connected to ground. CAUTION: To avoiddamaging the instrument, do not use it in a place where ambient temperature exceeds 40°C. CAUTION: To avoid damaging the instrument, do not input more than 15VDC to VCF. CAUTION: To avoid damaging the instrument, do not input more than 150VAC to Frequency Counter (FG3C only). CAUTION: To avoid damaging the instrument, do not input more than 10Vpp during EXT modulation operation (FG3C only). FOR UNITEDKINGDOM ONLY NOTE: This lead/appliance must only be wired by competent persons WARNING: THIS APPLIANCE MUST BE EARTHED IMPORTANT: The wires in this lead are coloured in accordance with the following code:
E N L
Green/Yellow Earth Blue Brown: Neutral Live (Phase)
As the colours of the wires in main leads may not correspond with the colours marking identified in your plug/appliance,proceed as follows: The wire which is coloured Green & Yellow must be connected to the Earth terminal marked with the letter E or by the earth symbol or coloured Green or Green & Yellow. The wire which is coloured Blue must be connected to the terminal which is marked with the letter N or coloured Blue or Black. The wire which is coloured Brown must be connected to the terminal marked with the...